ultrasonic sonicator generator probe for 20khz Ultrasonic Cell Disruptors
≥ 10 Pieces
If you have your own design, please contact us, there will have a professional customer service to communicate with you.
Don't have your files right now?No worries, Check out our art tips for more information.
If over 500pcs, mass production will start until pre-production sample has been approved.
Est. 7 days production
|warranty:||1 Year for Ultrasonic Homogenizer /mixer /cell disruptor|
|Place of Origin||CN;GUA|
|Brand Name||OURS ULTRASONIC|
|voltage:||220V/380V for Ultrasonic Homogenizer /mixer /cell disruptor|
|Certification:||CE ISO for Ultrasonic Homogenizer /mixer /cell disruptor|
|After Warranty Service||Video technical support|
|Local Service Location||none|
|Available Converter||250ul~500ml 1%~99% for Ultrasonic Homogenizer /mixer|
|Key Selling Points||Automatic|
|Power (w)||500W for Ultrasonic Homogenizer /mixer /cell disruptor|
|Output Amplitude||1/8 3/16 1/4 1/2 for Ultrasonic Homogenizer /mixer|
|After-sales Service Provided||No after-sales service|
|Barrel Volume (L)||0.25 L|
|Max. Loading Volume (L)||0.8 L|
|Max. Loading Capacity:||80ul~250ml for Ultrasonic Homogenizer /mixer /cell disruptor|
|Weight (KG)||35 KG|
|Power (kW)||0.5 kW|
Packaging & Delivery
- Selling Units:
- Single package size:
- 65 x 40 x 45 m
- Single gross weight:
- 45 kg
ultrasonic sonicator gene probe for 20khz Ultrasonic Cell Disruptors
The ultrasonic power supply generates high frequency electrical signals, which is applied to the piezoelectric crystals within the converter, where it is changed to mechanical vibrations. The longitudinal vibrations are amplified by the probe and transmitted to the cleaning liquids, which consist of alternate compressions and rarefactions. These pressure fluctuations create micro bubbles, which expand during the negative pressure excursions, and implode violently during the positive excursions. As the bubbles collapse, millions of shock waves, extreme in pressures and temperatures are generated at the implosion sites. These of the high pressure and temperature can be effectively used to obtain cleaning results.
The ultrasonic cleaning equipment can be applied to electronic components, silicon of the semiconductor, electronic board, optical lenses, audio magnetic head, electronic device, camera device, communication equipment, medical equipment, medical operation device, glass device, fine mechanical components, Biology HPLC, physics, chemistry, and medicine.
1. Precise and convenient digital displaying and controlling.
2. Digital display can show you the output power directly.
3. Output amplitude can be adjusted from 0% to 100%.
4. Two options available: Interval pulse modulate & constant output.
5. Pulse width and interval time can be separately set.
6. Its timer controlled and can switch off automatically when the set time is run off. You can turn it off when necessary.
7. Various kinds of converter are available.
Notice The foot switch and the buttons of the panel control the output of the instrument at the same time, after the foot switch insert in the plug. When the instrument is on work, please set the button (time and output amplitude) and outputfirstly, and then push the foot switch.
VI. Preparation before operation
Notice: If the instrument is put in the cold or hot environment for a long time, please wait for a moment till the room temperature.
1. Make sure the power button is “OFF”
2. Plug the power cable.
3. If other converters are needed, please change the “Tip” on the probe with the attached handle and rubber piece, and install the new converter to the probe.
4. Install the converter, probe parts on the lab support or in the enclosure.
5. Connect the converter cable to the connection unit.
6. Make sure the foot switch is connecting with the plug.
VII.How to use
100W,250W,500W,650W,800W,950W,1200W Connection of converter and probe:
Plug the power cable; make sure the power button is “ON”.
Set the max intensity though the buttons on the panel. The max intensity of the instrument is set at 30% max intensity at the factory. You can choose the proper max intensity.
If you want to change the instrument working mode, please press the “mode” button and then press “1/Y” button or “0/N” button for choosing the instrument working mode. And last please press the “Enter” button.
Please press the “Mode” button, and then please press the “Timer” button to set the total working time by press the number buttons on the front panel. Finally press the “Enter” button. If you want to change the time in the pulse working mode, please press the “Clear” button at fist. Second, please press the number buttons to set the time for continually working pulse mode and the time for pulse off. Finally press the “Enter” button.
If the converter has no ultrasonic vibration, please set the max intensity at 35% max intensity and adjust the tune bolt slowly.
Intensity of ultrasonic cleaner depends on the output of power. It is controlled through the intensity button on front panel. As the obstruction of the probe is strengthened, the power needed increases. Power device will detect this need and automatically increase the output power. Higher viscidity, larger diameter of tip or higher pressure will form greater obstruction to the probe. That will certainly increase power output. Turning the “Intensity button clockwise does not mean transporting greater power to the sample, only if when obstruction to the probe is large enough.
Do not turn on the switch when not connecting with the converter.
Do not let micro-oscillator vibrate over 10 seconds in the air.
Do not contact the vibrating probe with other articles except the sample.
No need register, just fill in contact info below, we will reply you within 24 hours!
Enter the code in the box below: